**Dynamic Characterization Testing (600 Hz)**

- 3 Axis testing frame
- In vertical axis (Z)
- Capability of sine sweep and sine dwell test
- Up to 600 Hz test frequency range capacity
- Up to ±25 kN rated force capacity with a load cell plate assembled by piezo-electric force trancducers for measuring and controlling
- Up to ±25 mm deflection for measuring and controlling
- Precisely controlling dynamic amplitude as based displacement or force
- Precisely controlling mean level as based displacement or force
- Defining the test frequencies as consecutively and arbitrarily in sine dwell tests
- Defining sweep rate as linear or logarithmic in sisne sweep tests
- Calculating frequency dependant loss, storage and complex dynamic rigidity, damping coefficent, phase angle, transmissibility, loss energy and etc.
- Correcting of dynamic displacement to be used stiffness calculation by taking measurement from an accelerometer on actuator after 120 Hz
- Compansating dynamic forces by subtracting inertial effect of total masses on load cell plate using an aditional accelerometer on that plate from total measured dynamic force
- Using a pre-load axis in horizontal dynamic test by force or deflection based control

- In horizontal axis (X)
- Capability of sine sweep and sine dwell test
- Up to 50 Hz test frequency range capacity
- Up to ±5 kN rated force capacity with a load cell plate assembled by piezo-electric force trancducers for measuring and controlling
- Up to ±25 mm deflection for measuring and controlling
- Precisely controlling dynamic amplitude as based displacement or force
- Precisely controlling mean level as based displacement or force
- Defining the test frequencies as consecutively and arbitrarily in sine dwell tests
- Defining sweep rate as linear or logarithmic in sisne sweep tests
- Calculating frequency dependant loss, storage and complex dynamic rigidity, damping coefficent, phase angle, transmissibility, loss energy and etc.
- Using a pre-load axis in vertical dynamic test by force or deflection based control

- In horizontal axis (Y)
- Manually adjustment in Y direction
- Capability of adjusting alignment of test specimen to upper mounting interface of test system by powerscrew handle
- Up to ±2 kN rated force capacity with a load cell plate assembled by piezo-electric force trancducers for measuring and monitoring not controlling

**Dynamic Characterization Testing (200 Hz)**

- Uni-axial testing frame
- Capability of sine dwell test
- Up to 200 Hz test frequency range (Max. Frequency capability depends to stiffness of specimens, )
- Up to ±50 kN rated force capacity
- Precisely controlling dynamic amplitude as based displacement or force
- Precisely controlling mean level as based displacement or force
- Defining the test frequencies as consecutivly and arbitrarily
- Calculating frequency dependant loss, storage and complex dynamic rigidity, damping coefficent, phase angle, transmissibility, loss energy and etc.

**High Frequency Dynamic Characterization Testing (2800 Hz)**

- Capability of sine sweep test
- Up to 2800 Hz test frequency range capacity
- Up to ±60 kN rated force capacity with a load cell plate assembled by piezo-electric force trancducers for measuring
- Up to ±3grms acceleration for measuring and controlling
- Precisely controlling dynamic amplitude as based acceleration
- Defining sweep rate as linear or logarithmic in sine sweep tests
- Calculating frequency dependant loss, storage and complex dynamic rigidity, damping coefficent, phase angle, transmissibility, loss energy and etc.